Overview of HP Agilent 1100 HPLC G1329A Autosampler
HP Agilent 1100 HPLC G1329A Autosampler is a thermostatted autosampler and weighs 31.3 lbs and has dimensions of (WxDxH) 13.5×17×8.1inches. The power consumption is 300 VA. The instrument requires an extra 2.5 cm (1.0 inch) of space on each side, and 8 cm (3.1 inches) at the rear to facilitate circulation of air and room for electrical connections.
HP Agilent 1100 HPLC G1329A Autosampler supports thermostattable Tray for 100 x 2 ml vials and the accessory kit contains tubing assembly, filter promo kit, CAN cable (1 m long), screw cap vials (clear 100/pk), blue screw caps(100/pk), label halftray, vial instruction sheet, wrenches (1/4 – 5/16 inch), Rheotool socket wrench (1/4 inch), Hex key (4 mm, 15 cm long, T-handle), Hex key (9/64 mm, 15 cm long, T- handle), Hex key ( 2.5 mm, 15 cm long, straight handle), ESD wrist strap, finger caps (pack of 15), Front door cooled autosampler (G1329-40301), air channel adapter (G1329-43200), cover insulation, capillary (0.17 mm, 900 mm G1329-87300), capillary heat exchanger.
Two status indicators are located on the front of HP Agilent 1100 HPLC G1329A Autosampler. The lower left indicates the power supply status and the upper right indicates the autosampler status. When the status indicator is OFF (but the power switch light is on), the instrument denotes a pre-run condition and is ready to begin an analysis. Green light as a status indicator indicates that the instrument is performing an analysis (which is run mode). A yellow indicators represents a not-ready condition. An error condition is indicated by red light which indicates that the instrument has detected an internal problem.
To maintain the quality of G1329A Autosampler you should routinely replace the rotor seal, check the stator, replace the needle and needle seat, check the ﬁnger caps, replace the metering unit seal, clean the piston, clean the solvent waste path, check the leak sensor, and perform a pressure test. The covers should be kept clean. HP Agilent 1100 HPLC G1329A Autosampler should be cleaned with a soft cloth slightly dampened with either water or a solution of water and a mild detergent. Avoid the use of an excessively damp cloth to prevent the liquid from dripping into the autosampler which could lead to a shock hazard or damage to the autosampler.
|GLP Features||Early maintenance feedback (EMF), electronic records of maintenance and errors|
|Communications||Controller-area network (CAN). GPIB (IEEE-448), RS232C, APG-remote standard, optional four external contact closures and BCD vial number output|
|Safety Features||Leak detection and safe leak handling, low voltages in maintenance areas, error detection and display|
|Injection Range||0.1 – 100 µl in 0.1 µl increments Up to 1500 µl with multiple draw (hardware modification
|Replicate Injections||1 – 99 from one vial|
|Precision||Typically < 0.5 % RSD of peak areas from 5 – 100 µl
Typically < 1 % RSD of peak areas from 1 – 5 µl
|Minimum Sample Volume||1 µl from 5 µl sample in 100 µl microvial, or 1 µl from 10 µl sample in 300 µl microvial|
|Carryover||Typically < 0.1 %, < 0.05 % with external needle cleaning|
|Sample Viscosity Range||0.2 – 50 cp|
|Replicate Injections Per Vial||1 – 99|
|Sample Capacity||100 × 2-ml vials in 1 tray
40 × 2-ml vials in ½ tray
15 × 6-ml vials in ½ tray (Agilent vials only)
|Injection Cycle Time||Typically 50 s depending on draw speed and injection volume|
HP Agilent 1100 HPLC G1329A Autosampler Manual
Laboratory Controls offers a best in the business full 90 day warranty on all refurbished items. This warranty covers your part against any material and or craftsmanship issues within one full year from purchase. Laboratory Controls will repair or replace your part at no charge. Please contact us for further information if you are in need of a repair.